CHE301 Assignment 2 Solution and Discussion
Assignment No. 2
CHE301 – Analytical Chemistry & Instrumentaions
Total Marks: 10
Due date: 16-01-2020
• Make sure that you upload the solution file before due date. No assignment will be accepted through e-mail after the due date.
• Use the font style “Times New Roman” and font size “12”.
• Compose your document in MS-Word, any file created in any other format will not be accepted and assigned zero mark.
• Use black and blue font colors only.
• To solve this assignment, you should have good command over lectures 64-74.
• This is not a group assignment, it is an individual assignment so be careful and avoid copying others’ work
• Plagiarism of assignment file (MS Word) will be checked. There will be negative marking in case of plagiarism detected.
• Give the answer according to question only and avoid irrelevant details.
Please note that your assignment will not be graded if:
• It is submitted after due date
• The file you uploaded does not open
• The file you uploaded is copied from someone else
• It is in format other than .doc
• Cheating or copying of assignment is strictly prohibited. The cheated or copied assignment will be marked ‘Zero’.
Question No. 1
Discuss Instrumentation & working of HPLC? (10)
@Arbaz-Kaleem Solution idea posted
As shown in the schematic diagram in Figure above, HPLC instrumentation includes a pump, injector, column, detector and integrator or acquisition and display system. The heart of the system is the column where separation occurs.
- Solvent Resorvoir
Mobile phase contents are contained in a glass resorvoir. The mobile phase, or solvent, in HPLC is usually a mixture of polar and non-polar liquid components whose respective concentrations are varied depending on the composition of the sample.
A pump aspirates the mobile phase from the solvent resorvoir and forces it through the system’s column and detecter. Depending on a number of factors including column dimensions, particle size of the stationary phase, the flow rate and composition of the mobile phase, operating pressures of up to 42000 kPa (about 6000 psi) can be generated.
3. Sample Injector
The injector can be a single injection or an automated injection system. An injector for an HPLC system should provide injection of the liquid sample within the range of 0.1-100 mL of volume with high reproducibility and under high pressure (up to 4000 psi).
Columns are usually made of polished stainless steel, are between 50 and 300 mm long and have an internal diameter of between 2 and 5 mm. They are commonly filled with a stationary phase with a particle size of 3–10 µm.
Columns with internal diameters of less than 2 mm are often referred to as microbore columns. Ideally the temperature of the mobile phase and the column should be kept constant during an analysis.
The HPLC detector, located at the end of the column detect the analytes as they elute from the chromatographic column. Commonly used detectors are UV-spectroscopy, fluorescence, mass-spectrometric and electrochemical detectors.
6. Data Collection Devices
Signals from the detector may be collected on chart recorders or electronic integrators that vary in complexity and in their ability to process, store and reprocess chromatographic data. The computer integrates the response of the detector to each component and places it into a chromatograph that is easy to read and interpret.
Discuss Instrumentation & working of HPLC?
Arbaz Kaleem last edited by
@zareen assignment solution plz
![alt text](image url)